- Raman spectroscopy
Raman spectroscopy ( //; named after C. V. Raman) is a spectroscopic technique used to study vibrational, rotational, and other low-frequency modes in a system. It relies on inelastic scattering, or Raman scattering, of monochromatic light, usually from a laser in the visible, near infrared, or near ultraviolet range. The laser light interacts with molecular vibrations, phonons or other excitations in the system, resulting in the energy of the laser photons being shifted up or down. The shift in energy gives information about the vibrational modes in the system. Infrared spectroscopy yields similar, but complementary, information.
Typically, a sample is illuminated with a laser beam. Light from the illuminated spot is collected with a lens and sent through a monochromator. Wavelengths close to the laser line, due to elastic Rayleigh scattering, are filtered out while the rest of the collected light is dispersed onto a detector.
Spontaneous Raman scattering is typically very weak, and as a result the main difficulty of Raman spectroscopy is separating the weak inelastically scattered light from the intense Rayleigh scattered laser light. Historically, Raman spectrometers used holographic gratings and multiple dispersion stages to achieve a high degree of laser rejection. In the past, photomultipliers were the detectors of choice for dispersive Raman setups, which resulted in long acquisition times. However, modern instrumentation almost universally employs notch or edge filters for laser rejection and spectrographs (either axial transmissive (AT), Czerny-Turner (CT) monochromator, or FT (Fourier transform spectroscopy based), and CCD detectors.
There are a number of advanced types of Raman spectroscopy, including surface-enhanced Raman, resonance Raman, tip-enhanced Raman, polarised Raman, stimulated Raman (analogous to stimulated emission), transmission Raman, spatially-offset Raman, and hyper Raman.
The Raman effect occurs when light impinges upon a molecule and interacts with the electron cloud and the bonds of that molecule. For the spontaneous Raman effect, which is a form of light scattering, a photon excites the molecule from the ground state to a virtual energy state. When the molecule relaxes it emits a photon and it returns to a different rotational or vibrational state. The difference in energy between the original state and this new state leads to a shift in the emitted photon's frequency away from the excitation wavelength. The Raman effect, which is a light scattering phenomenon, should not be confused with absorption (as with fluorescence) where the molecule is excited to a discrete (not virtual) energy level.
If the final vibrational state of the molecule is more energetic than the initial state, then the emitted photon will be shifted to a lower frequency in order for the total energy of the system to remain balanced. This shift in frequency is designated as a Stokes shift. If the final vibrational state is less energetic than the initial state, then the emitted photon will be shifted to a higher frequency, and this is designated as an Anti-Stokes shift. Raman scattering is an example of inelastic scattering because of the energy transfer between the photons and the molecules during their interaction.
A change in the molecular polarization potential — or amount of deformation of the electron cloud — with respect to the vibrational coordinate is required for a molecule to exhibit a Raman effect. The amount of the polarizability change will determine the Raman scattering intensity. The pattern of shifted frequencies is determined by the rotational and vibrational states of the sample.
Although the inelastic scattering of light was predicted by Adolf Smekal in 1921, it is not until 1928 that it was observed in practice. The Raman effect was named after one of its discoverers, the Indian scientist Sir C. V. Raman who observed the effect by means of sunlight (1928, together with K. S. Krishnan and independently by Grigory Landsberg and Leonid Mandelstam). Raman won the Nobel Prize in Physics in 1930 for this discovery accomplished using sunlight, a narrow band photographic filter to create monochromatic light, and a "crossed filter" to block this monochromatic light. He found that a small amount of light had changed frequency and passed through the "crossed" filter.
Systematic pioneering theory of the Raman effect was developed by Czechoslovak physicist George Placzek between 1930 and 1934. The mercury arc became the principal light source, first with photographic detection and then with spectrophotometric detection. At the present time, lasers are used as light sources.
Raman shift are typically expressed in wavenumbers, which have units of inverse length. In order to convert between spectral wavelength and wavenumbers of shift in the Raman spectrum, the following formula can be used:
where Δw is the Raman shift expressed in wavenumber, λ0 is the excitation wavelength, and λ1 is the Raman spectrum wavelength. Most commonly, the units chosen for expressing wavenumber in Raman spectra is inverse centimeters (cm−1). Since wavelength is often expressed in units of nanometers (nm), the formula above can scale for this units conversion explicitly, giving
Raman spectroscopy is commonly used in chemistry, since vibrational information is specific to the chemical bonds and symmetry of molecules. Therefore, it provides a fingerprint by which the molecule can be identified. For instance, the vibrational frequencies of SiO, Si2O2, and Si3O3 were identified and assigned on the basis of normal coordinate analyses using infrared and Raman spectra. The fingerprint region of organic molecules is in the (wavenumber) range 500–2000 cm−1. Another way that the technique is used is to study changes in chemical bonding, as when a substrate is added to an enzyme.
Raman gas analyzers have many practical applications. For instance, they are used in medicine for real-time monitoring of anaesthetic and respiratory gas mixtures during surgery.
In solid-state physics, spontaneous Raman spectroscopy is used to, among other things, characterize materials, measure temperature, and find the crystallographic orientation of a sample. As with single molecules, a given solid material has characteristic phonon modes that can help an experimenter identify it. In addition, Raman spectroscopy can be used to observe other low frequency excitations of the solid, such as plasmons, magnons, and superconducting gap excitations. The spontaneous Raman signal gives information on the population of a given phonon mode in the ratio between the Stokes (downshifted) intensity and anti-Stokes (upshifted) intensity.
Raman scattering by an anisotropic crystal gives information on the crystal orientation. The polarization of the Raman scattered light with respect to the crystal and the polarization of the laser light can be used to find the orientation of the crystal, if the crystal structure (to be specific, its point group) is known.
Raman active fibers, such as aramid and carbon, have vibrational modes that show a shift in Raman frequency with applied stress. Polypropylene fibers also exhibit similar shifts. The radial breathing mode is a commonly used technique to evaluate the diameter of carbon nanotubes. In nanotechnology, a Raman microscope can be used to analyze nanowires to better understand the composition of the structures.
Spatially-offset Raman spectroscopy (SORS), which is less sensitive to surface layers than conventional Raman, can be used to discover counterfeit drugs without opening their internal packaging, and for non-invasive monitoring of biological tissue. Raman spectroscopy can be used to investigate the chemical composition of historical documents such as the Book of Kells and contribute to knowledge of the social and economic conditions at the time the documents were produced. This is especially helpful because Raman spectroscopy offers a non-invasive way to determine the best course of preservation or conservation treatment for such materials.
Raman spectroscopy has also been used to confirm the prediction of existence of low-frequency phonons  in proteins and DNA (see, e.g.,     greatly stimulating the studies of low-frequency collective motion in proteins and DNA and their biological functions.
Raman spectroscopy offers several advantages for microscopic analysis. Since it is a scattering technique, specimens do not need to be fixed or sectioned. Raman spectra can be collected from a very small volume (< 1 µm in diameter); these spectra allow the identification of species present in that volume. Water does not generally interfere with Raman spectral analysis. Thus, Raman spectroscopy is suitable for the microscopic examination of minerals, materials such as polymers and ceramics, cells, proteins and forensic trace evidence. A Raman microscope begins with a standard optical microscope, and adds an excitation laser, a monochromator, and a sensitive detector (such as a charge-coupled device (CCD), or photomultiplier tube (PMT)). FT-Raman has also been used with microscopes. Ultraviolet microscopes and UV enhanced optics must be used when a UV laser source is used for Raman microspectroscopy.
In direct imaging, the whole field of view is examined for scattering over a small range of wavenumbers (Raman shifts). For instance, a wavenumber characteristic for cholesterol could be used to record the distribution of cholesterol within a cell culture.
The other approach is hyperspectral imaging or chemical imaging, in which thousands of Raman spectra are acquired from all over the field of view. The data can then be used to generate images showing the location and amount of different components. Taking the cell culture example, a hyperspectral image could show the distribution of cholesterol, as well as proteins, nucleic acids, and fatty acids. Sophisticated signal- and image-processing techniques can be used to ignore the presence of water, culture media, buffers, and other interferents.
Raman microscopy, and in particular confocal microscopy, has very high spatial resolution. For example, the lateral and depth resolutions were 250 nm and 1.7 µm, respectively, using a confocal Raman microspectrometer with the 632.8 nm line from a Helium-Neon laser with a pinhole of 100 µm diameter. Since the objective lenses of microscopes focus the laser beam to several micrometres in diameter, the resulting photon flux is much higher than achieved in conventional Raman setups. This has the added benefit of enhanced fluorescence quenching. However, the high photon flux can also cause sample degradation, and for this reason some setups require a thermally conducting substrate (which acts as a heat sink) in order to mitigate this process.
By using Raman microspectroscopy, in vivo time- and space-resolved Raman spectra of microscopic regions of samples can be measured. As a result, the fluorescence of water, media, and buffers can be removed. Consequently in vivo time- and space-resolved Raman spectroscopy is suitable to examine proteins, cells and organs.
Raman microscopy for biological and medical specimens generally uses near-infrared (NIR) lasers (785 nm diodes and 1064 nm Nd:YAG are especially common). This reduces the risk of damaging the specimen by applying higher energy wavelengths. However, the intensity of NIR Raman is low (owing to the ω4 dependence of Raman scattering intensity), and most detectors required very long collection times. Recently, more sensitive detectors have become available, making the technique better suited to general use. Raman microscopy of inorganic specimens, such as rocks and ceramics and polymers, can use a broader range of excitation wavelengths.
The polarization of the Raman scattered light also contains useful information. This property can be measured using (plane) polarized laser excitation and a polarization analyzer. Spectra acquired with the analyzer set at both perpendicular and parallel to the excitation plane can be used to calculate the depolarization ratio. Study of the technique is useful in teaching the connections between group theory, symmetry, Raman activity, and peaks in the corresponding Raman spectra.
The spectral information arising from this analysis gives insight into molecular orientation and vibrational symmetry. In essence, it allows the user to obtain valuable information relating to the molecular shape, for example in synthetic chemistry or polymorph analysis. It is often used to understand macromolecular orientation in crystal lattices, liquid crystals or polymer samples.
Several variations of Raman spectroscopy have been developed. The usual purpose is to enhance the sensitivity (e.g., surface-enhanced Raman), to improve the spatial resolution (Raman microscopy), or to acquire very specific information (resonance Raman).
- Surface Enhanced Raman Spectroscopy (SERS) - Normally done in a silver or gold colloid or a substrate containing silver or gold. Surface plasmons of silver and gold are excited by the laser, resulting in an increase in the electric fields surrounding the metal. Given that Raman intensities are proportional to the electric field, there is large increase in the measured signal (by up to 1011). This effect was originally observed by Martin Fleischmann but the prevailing explanation was proposed by Van Duyne in 1977. A comprehensive theory of the effect was given by Lombardi and Birke.
- Resonance Raman spectroscopy - The excitation wavelength is matched to an electronic transition of the molecule or crystal, so that vibrational modes associated with the excited electronic state are greatly enhanced. This is useful for studying large molecules such as polypeptides, which might show hundreds of bands in "conventional" Raman spectra. It is also useful for associating normal modes with their observed frequency shifts.
- Surface-Enhanced Resonance Raman Spectroscopy (SERRS) - A combination of SERS and resonance Raman spectroscopy that uses proximity to a surface to increase Raman intensity, and excitation wavelength matched to the maximum absorbance of the molecule being analysed.
- Angle Resolved Raman Spectroscopy - Not only are standard Raman results recorded but also the angle with respect to the incident laser. If the orientation of the sample is known then detailed information about the phonon dispersion relation can also be gleamed from a single test.
- Hyper Raman - A non-linear effect in which the vibrational modes interact with the second harmonic of the excitation beam. This requires very high power, but allows the observation of vibrational modes that are normally "silent". It frequently relies on SERS-type enhancement to boost the sensitivity.
- Spontaneous Raman Spectroscopy (SRS) - Used to study the temperature dependence of the Raman spectra of molecules.
- Optical Tweezers Raman Spectroscopy (OTRS) - Used to study individual particles, and even biochemical processes in single cells trapped by optical tweezers.
- Stimulated Raman Spectroscopy - A spatially coincident, two color pulse (with polarization either parallel or perpendicular) transfers the population from ground to a rovibrationally excited state, if the difference in energy corresponds to an allowed Raman transition, and if neither frequency corresponds to an electronic resonance. Two photon UV ionization, applied after the population transfer but before relaxation, allows the intra-molecular or inter-molecular Raman spectrum of a gas or molecular cluster (indeed, a given conformation of molecular cluster) to be collected. This is a useful molecular dynamics technique.
- Spatially Offset Raman Spectroscopy (SORS) - The Raman scatter is collected from regions laterally offset away from the excitation laser spot, leading to significantly lower contributions from the surface layer than with traditional Raman spectroscopy.
- Coherent anti-Stokes Raman spectroscopy (CARS) - Two laser beams are used to generate a coherent anti-Stokes frequency beam, which can be enhanced by resonance.
- Raman optical activity (ROA) - Measures vibrational optical activity by means of a small difference in the intensity of Raman scattering from chiral molecules in right- and left-circularly polarized incident light or, equivalently, a small circularly polarized component in the scattered light.
- Transmission Raman - Allows probing of a significant bulk of a turbid material, such as powders, capsules, living tissue, etc. It was largely ignored following investigations in the late 1960s but was rediscovered in 2006 as a means of rapid assay of pharmaceutical dosage forms. There are also medical diagnostic applications.
- Inverse Raman spectroscopy.
- Tip-Enhanced Raman Spectroscopy (TERS) - Uses a metallic (usually silver-/gold-coated AFM or STM) tip to enhance the Raman signals of molecules situated in its vicinity. The spatial resolution is approximately the size of the tip apex (20-30 nm). TERS has been shown to have sensitivity down to the single molecule level. [reference?]
- ^ a b Gardiner, D.J. (1989). Practical Raman spectroscopy. Springer-Verlag. ISBN 978-0387502540.
- ^ Placzek G.: "Rayleigh Streeung und Raman Effekt", In: Hdb. der Radiologie, Vol. VI., 2, 1934, p. 209
- ^ Khanna, R.K. (1981). "Raman-spectroscopy of oligomeric SiO species isolated in solid methane". Journal of Chemical Physics 74 (4): 2108. doi:10.1063/1.441393.
- ^ "Fake drugs caught inside the pack". BBC News. 2007-01-31. http://news.bbc.co.uk/2/hi/health/6314287.stm. Retrieved 2008-12-08.
- ^ Irish classic is still a hit (in calfskin, not paperback) - New York Times, nytimes.com
- ^ Ben Vogel (29 August 2008). "Raman spectroscopy portends well for standoff explosives detection". Jane's. http://www.janes.com/news/transport/business/jar/jar080829_1_n.shtml. Retrieved 2008-08-29.
- ^ Kuo-Chen Chou and Nian-Yi Chen (1977) The biological functions of low-frequency phonons. Scientia Sinica, 20, 447-457.
- ^ Urabe, H., Tominaga, Y. and Kubota, K. (1983) Experimental evidence of collective vibrations in DNA double helix Raman spectroscopy. Journal of Chemical Physics, 78, 5937-5939.
- ^ Chou, K.C. (1983) Identification of low-frequency modes in protein molecules. Biochemical Journal, 215, 465-469.
- ^ Chou, K.C. (1984) Low-frequency vibration of DNA molecules. Biochemical Journal, 221, 27-31.
- ^ Urabe, H., Sugawara, Y., Ataka, M. and Rupprecht, A. (1998) Low-frequency Raman spectra of lysozyme crystals and oriented DNA films: dynamics of crystal water. Biophys J, 74, 1533-1540.
- ^ Kuo-Chen Chou (1988) Review: Low-frequency collective motion in biomacromolecules and its biological functions. Biophysical Chemistry, 30, 3-48.
- ^ Chou, K.C. (1989) Low-frequency resonance and cooperativity of hemoglobin. Trends in Biochemical Sciences, 14, 212.
- ^ Ellis DI, Goodacre R (August 2006). "Metabolic fingerprinting in disease diagnosis: biomedical applications of infrared and Raman spectroscopy". Analyst 131 (8): 875–85. doi:10.1039/b602376m. PMID 17028718.
- ^ Khanna, R.K. (1957). Evidence of ion-pairing in the polarized Raman spectra of a Ba2+CrO doped KI single crystal. John Wiley & Sons, Ltd. doi:10.1002/jrs.1250040104.
- ^ Jeanmaire DL, van Duyne RP (1977). "Surface Raman Electrochemistry Part I. Heterocyclic, Aromatic and Aliphatic Amines Adsorbed on the Anodized Silver Electrode". Journal of Electroanalytical Chemistry (Elsevier Sequouia S.A.) 84: 1–20. doi:10.1016/S0022-0728(77)80224-6.
- ^ Lombardi JR, Birke RL (2008). "A Unified Approach to Surface-Enhanced Raman Spectroscopy". [Journal of Physical Chemistry C] (American Chemical Society) 112: 5605–5617. doi:10.1021/jp800167+CCC.
- ^ Chao RS, Khanna RK, Lippincott ER (1974). "Theoretical and experimental resonance Raman intensities for the manganate ion". J Raman Spectroscopy 3 (2-3): 121. doi:10.1002/jrs.1250030203.
- ^ Zachary J. Smith and Andrew J. Berger (2008). "Integrated Raman- and angular-scattering microscopy". Opt. Lett. 3 (7): 714-716. doi:10.1364/OL.33.000714.
- ^ Kneipp K, et al. (1999). "Surface-Enhanced Non-Linear Raman Scattering at the Single Molecule Level". Chem. Phys. 247: 155–162. doi:10.1016/S0301-0104(99)00165-2.
- ^ Matousek P, Clark IP, Draper ERC, et al. (2005). "Subsurface Probing in Diffusely Scattering Media using Spatially Offset Raman Spectroscopy". Applied Spectroscopy 59 (12): 393. doi:10.1366/000370205775142548. PMID 16390587.
- ^ Barron LD, Hecht L, McColl IH, Blanch EW (2004). "Raman optical activity comes of age". Molec. Phys. 102 (8): 731–744. doi:10.1080/00268970410001704399.
- ^ B. Schrader, G. Bergmann, Fresenius. Z. (1967). Anal. Chem.: 225–230.
- ^ P. Matousek, A. W. Parker (2006). "Bulk Raman Analysis of Pharmaceutical Tablets". Applied Spectroscopy 60 (12): 1353–1357. doi:10.1366/000370206779321463. PMID 17217583.
- ^ P. Matousek, N. Stone (2007). "Prospects for the diagnosis of breast cancer by noninvasive probing of calcifications using transmission Raman spectroscopy". Journal of Biomedical Optics 12 (2): 024008. doi:10.1117/1.2718934. PMID 17477723.
- Raman on SiGe Superlattice, princetoninstruments.com
- An introduction on Raman Scattering, d3technologies.co.uk
- Raman Spectroscopy Applications, renishaw.com
- Raman Data Search and Storage - The free application with a wonderful database of Raman data (vibrations, assignment) with storage function and Raman spectra (discussions) with search function. ramandata.sourceforge.net
- Romanian Database of Raman Spectroscopy - This database contains mineral species (natural and synthetic) with description of crystal structure, sample image, number of sample, origin, Raman spectrum and vibrations, Raman discussion and references. Also, this site contains artefacts sample with sample image and pigment spectrum; black, red, white or blue pigment. rdrs.uaic.ro
- Chemical Imaging Without Dyeing, witec.de
- DoITPoMS Teaching and Learning Package - Raman Spectroscopy - an introduction, aimed at undergraduate level. doitpoms.ac.uk
- Raman Spectroscopy Tutorial - A detailed explanation of Raman Spectroscopy including Resonance-Enhanced Raman Scattering and Surface-Enhanced Raman Scattering. 188.8.131.52
- The Science Show, ABC Radio National - Interview with Scientist on NASA funded project to build Raman Spectrometer for the 2009 Mars mission: a cellular phone size device to detect almost any substance known, with commercial <USD$5000 commercial spin-off, prototyped by June 2006. abc.net.au/rn
- Raman spectroscopy for medical diagnosis by J. Griffiths from the June 1, 2007 issue of Analytical Chemistry, pubs.acs.org
- Spontaneous Raman Scattering (SRS), lavision.de
- Painless laser device could spot early signs of disease, BBC News, 2010-09-26
Wikimedia Foundation. 2010.
Look at other dictionaries:
Raman spectroscopy — n a spectroscopic technique in which the Raman spectrum of a substance is analyzed to determine the properties (as the structure) of the substance … Medical dictionary
Raman Spectroscopy — Raman Spectroscopy Рамановская спектроскопия (спектроскопия комбинационного рассеяния) Раздел оптической спектроскопии, изучающий взаимодействие монохроматического излучения с веществом, сопровождающееся изменением энергии рассеянного… … Толковый англо-русский словарь по нанотехнологии. - М.
Raman spectroscopy — Ramano spektroskopija statusas T sritis Standartizacija ir metrologija apibrėžtis Medžiagoje išsklaidytos vienspalvės šviesos spektrų tyrimas. atitikmenys: angl. Raman spectroscopy vok. Raman Spektroskopie, f rus. рамановская спектроскопия, f;… … Penkiakalbis aiškinamasis metrologijos terminų žodynas
Raman spectroscopy — Ramano spektroskopija statusas T sritis Standartizacija ir metrologija apibrėžtis Optinės spektroskopijos skyrius, nagrinėjantis Ramano sklaidą medžiagose. atitikmenys: angl. Raman spectroscopy vok. Raman Spektroskopie, f rus. рамановская… … Penkiakalbis aiškinamasis metrologijos terminų žodynas
Raman spectroscopy — Ramano spektroskopija statusas T sritis fizika atitikmenys: angl. Raman spectroscopy vok. Raman Spektroskopie, f rus. рамановская спектроскопия, f; спектроскопия комбинационного рассеяния, f pranc. spectroscopie Raman, f … Fizikos terminų žodynas
Raman spectroscopy — Ramano spektroskopija statusas T sritis chemija apibrėžtis Optinės spektroskopijos skyrius, nagrinėjantis Ramano sklaidą medžiagose. atitikmenys: angl. Raman spectroscopy rus. Рамановская спектроскопия; спектроскопия комбинационного рассеяния… … Chemijos terminų aiškinamasis žodynas
Raman spectroscopy — noun A form of spectroscopy that uses the Raman effect to study low frequency vibrational and rotational modes of molecules … Wiktionary
Raman spectroscopy — Method for measuring the Raman spectrum, the plot of Raman scattering of light that produces weak radiation at frequencies not present in the incident radiation. The spectrum is characteristic of the compound and independent of the wavelength of… … Dictionary of molecular biology
raman spectroscopy — noun Usage: usually capitalized R : a spectroscopic technique in which the Raman spectrum of a substance is analyzed to determine the properties (as the structure) of the substance … Useful english dictionary
Resonance Raman spectroscopy — Resonance Raman (RR) spectroscopy is a specialized implementation of the more general Raman spectroscopy. Overview As in Raman spectroscopy, RR spectroscopy provides information about the vibrations of molecules, and can also be used for… … Wikipedia