- Bag valve mask
A bag valve mask (also known as a BVM or Ambu bag) is a hand-held device used to provide
ventilationto a patient who is not breathing or who is breathing inadequately. The device is a normal part of a resuscitationkit for trained professional, such as ambulancecrew. The BVM is frequently used in hospitals, and is an essential part of a crash cart. The device is used extensively in the operating room to ventilate an anaesthetised patient in the minutes before a mechanical ventilator is attached. The device is self-filling with air, although additional oxygen(O2) can be added.
Use of the BVM to ventilate a patient is frequently called "bagging" the patient.Daniel Limmer and Michael F. O'Keefe. 2005. "Emergency Care" 10th ed. Edward T. Dickinson, Ed. Pearson, Prentice Hall. Upper Saddle River, New Jersey. Page 140.] Bagging is regularly necessary in medical emergencies when the patient's breathing is insufficient (
respiratory failure) or has ceased completely ( respiratory arrest). The BVM resuscitator is used in order to manually provide " mechanical ventilation" in preference to mouth-to-mouth resuscitation (either direct or through an adjunct such as a pocket mask).
right|thumb|The Ambu Resuscitator bag or BVM, this version shows is a hybrid MkIII body, Mark IV head and with an old obsolete latex inflatable seal on the mask. The part labelled 1 is a flexible mask designed to seal to the patient's face, and the part labelled 3 is a self-filling bag, i.e. re-fills as an action of the elastic re-coil of the bag after compression.
The BVM consists of a flexible air chamber, about the size of a rugby ball, attached to a face mask via a shutter valve. When the air chamber or "bag" is squeezed, the device forces air through into the patient's lungs; when the bag is released, it self-inflates, drawing in ambient air or a low pressure oxygen flow supplied from a regulated cylinder, whilst the patient's lungs deflate to the air through the one way valve.
Bag and valve combinations can also be attached to an alternate airway adjunct, such as an endo-tracheal tube or
laryngeal mask airway, instead of the mask with which it is supplied. Often a small HME filter is also used (Heat & Moisture exchanger or simple put - a Humidifying / bacterial filter - see diagram on the right)
A bag valve mask can be used without being attached to an oxygen tank to provide air to the patient, but supplemental oxygen is recommended since it increases the partial pressure of oxygen in the air, helping to increase perfusion in the patient.
Most devices also have a reservoir which can fill with oxygen while the patient is exhaling (a process which happens passively), in order to increase the amount of oxygen that can be delivered to the patient to nearly 100%. [cite book|last=Stoy|first=Walt|title=Mosby's EMT-Basic Textbook|publisher=Mosby/JEMS|date=2004|ISBN=0323034381|url=http://www3.us.elsevierhealth.com/promo/Stoy/pdfs/ch8.pdf]
Bag valve masks come in different sizes to fit infants, children, and adults.
Most types of the device are now disposable and therefore single use, while others are designed to be cleaned and reused. Many users have moved away from the reusable bags to reduce the possibility of cross-contamination.
Method of operation
The BVM directs the gas inside it via a one-way
valvewhen compressed by a rescuer; the gas is then delivered through a mask and into the patient's trachea, bronchus and into the lungs. In order to be effective, a bag valve mask must deliver between 500 and 800 milliliters of air to the patient's lungs, but if oxygen is provided through the tubing and if the patient's chest rises with each inhalation (indicating that adequate amounts of air are reaching the lungs), 400 to 600 ml may still be adequate. Squeezing the bag once every 5 seconds for an adult or once every 3 seconds for an infant or child provides an adequate respiratory rate(12 respirations per minute in an adult and 20 per minute in a child or infant). ["Emergency Care", Pages 142-3]
Professional rescuers are taught to ensure that the mask portion of the BVM is properly sealed around the patient's face (that is, to ensure proper "mask seal"); otherwise, air escapes from the mask and is not pushed into the lungs. In order to maintain this protocol, some protocols use a method of ventilation involving two rescuers: one rescuer to hold the mask to the patient's face with both hands and ensure a mask seal, while the other squeezes the bag. ["Emergency Care", Page 141.] However, as most
ambulances have only two members of crew, the other crew member is likely to be doing compressions in the case of CPR, or may be performing other skills such as cannulation. In this case, or if no other options are available, the BVM can also be operated by a single rescuer who holds the mask to the patient's face with one hand, in the anaesthetists grip, and squeezes the bag with the other.
When using a BVM, as with other methods of insufflation, there is a risk of over-inflating the lungs. This can lead to pressure damage to the lungs themselves, and can also cause air to enter the stomach, causing
gastric distentionwhich can make it more difficult to inflate the lungs and which can cause the patient to vomit. This can be avoided by care on behalf of the rescuer. Alternatively, some models of BVM (usually Paediatric) are fitted with a valve which prevents over inflation, by venting the pressure when a pre-set pressure is reached.
endotracheal tube(ETT) can be inserted by a trained practitioner and can substitute for the mask portion of the BVM. This provides a more secure fit and is easier to manage during emergency transport, since the ET tube is sealed with an inflatable cuff in the trachea, so that any regurgitation cannot enter the lungs. Such material can severely damage the lung tissue, and in the absence of an ET tube, could choke the patient by obstructing the airway. Inhalation of stomach contents can be fatal; the after effects can cause Mendelson's syndromeor aspiration pneumonia.
Some rescuers may also choose to use a different form of resuscitation adjunt, such as an
oropharyngeal airwayor Laryngeal mask airway, which would be inserted and then used with the BVM.
In a hospital, long-term mechanical ventilation is provided by using more complex devices such as an intensive care ventilator, rather than by a BVM, which requires at least one person to operate it constantly.
flow-restricted, oxygen-powered ventilation device(FROPVD) is similar to a BVM in that oxygen is pushed through a mask into the patient's lungs, but unlike a BVM, in the FROPVD the pressure needed to push air into the patient's lungs is generated by oxygen via a pressure regulator from a cylinder rather than by squeezing a bag.
One proprietary brand of a self-inflating BVM resuscitator is called the Ambu bag. The concept for the original Ambu bag was developed in 1953 by the German engineer, Dr. Holger Hesse, and his partner, Danish anaesthetist, Henning Ruben. In 1956, the world's first non-electric, self-inflating resuscitator was ready for production by their company, Ambu A/S, which still produces a wide range of single-patient and multi-use resuscitators.
* [http://vam.anest.ufl.edu/checkout/check-sirb.html A free transparent reality simulation of the self-inflating manual resuscitator ("Ambu" bag)]
Wikimedia Foundation. 2010.
Look at other dictionaries:
bag valve mask — noun A bag, used to resuscitate a patient by supplying air or oxygen directly to the trachea and into the lungs … Wiktionary
Bag mask valve — #REDIRECT Bag valve mask … Wikipedia
AMBU BAG — Bag valve mask resusicitator used in an Ambulance (Medical » Hospitals) … Abbreviations dictionary
Non-rebreather mask — A non rebreather mask, or NRB, is a device used in medical emergencies that require oxygen therapy. An NRB requires that the patient can breathe unassisted, but unlike low flow nasal cannula, the NRB allows for the delivery of higher… … Wikipedia
Pocket mask — A pocket mask, or pocket face mask or CPR mask, is a device used to safely deliver rescue breaths during a cardiac arrest or respiratory arrest.DesignThe pocket mask is a small device that can be carried on one s person. Air is administered to… … Wikipedia
Positive end-expiratory pressure valve — Description = A PEEP Valve, or Positive end expiratory pressure valve is a device used for Continuous Positive Airway Pressure(CPAP), Biphasic Positive Airway Pressure (also called Bilevel Positive Airway Pressure) or an intubated patient. On an… … Wikipedia
Oxygen mask — A plastic oxygen mask on an ER patient. An oxygen mask provides a method to transfer breathing oxygen gas from a storage tank to the lungs. Oxygen masks may cover the nose and mouth (oral nasal mask) or the entire face (full face mask). They may… … Wikipedia
BLB mask — a face mask that has a combined inspiratory and expiratory valve and a bag for rebreathing, used mainly with oxygen delivery systems at high altitudes, and occasionally for clinical administration of oxygen … Medical dictionary
BVM — Bag Valve Mask (Medical » Physiology) * Bag Valve mask (for ventilating a patient) (Medical » Physiology) … Abbreviations dictionary
BVM — bag valve mask; bronchovascular markings; Bureau of Veterinary Medicine … Medical dictionary